The Axon System is designed for posterior stabilization of the upper spine taking into account the variations of patient anatomy. A posterior fusion and stabilization procedure is often used to treat instabilities secondary to traumatic injury, rheumatoid arthritis, ankylosing spondylitis, neoplastic disease, infections and previous laminectomy. The Axon System is based on the CerviFix/StarLock System. This allows an extension of a construct from the occiput to the lower spine using the Universal Spine System (USS).
The top-loading screws have a poly-axial head allowing an angulation of 30 based on the locking system of the Variable Axis Screw System. The self-tapping screws are available as 3.5 and 4.0 mm cancellous screws from 826 mm thread lengths and as 3.5 mm cortex screws from 2850 mm lengths. They can be used as lateral mass and transpedicular screw.
3.5 mm and 4.0 mm screws (2850 mm thread lengths)
Additional lengths of the 3.5 and 4.0 mm Axon screws in cancellous thread profile are available. The system now provides screws from 8 to 50 mm lengths in 2 mm increments to accommodate varying patient anatomy and different surgical techniques.
The Transverse Bar eases construct assembly if the screws are not inserted in line due to anatomical requirements. It provides a lateral offset of up to 9 mm from the 3.5 mm rod to the screw. It eliminates the need for severe rod contouring.The preassembled Transconnector to lock the assembly is placed after the Axon construct is in position. Bushings allow the clamps to be placed offset to each other.
The improved hooks for sublaminar insertion offer a wider hook opening. The offset between rod and hooks has been reduced. The instruments of the CerviFix set has been redesigned featuring an Adjustable Drill Guide, Pedicle Preparation Set, Bone-Screwdriver with Holding Sleeve, Alignment Tool, Torque Limiting Handle for Locking Screw, Rod Pusher/Countertorque, and Distraction and Compression Forceps.
2.4 mm K-wire with stop
The 2.4 mm K-wire with stop is for surgeons who prefer a K-wire approach instead of the standard drill bit procedure.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
"approved by AO Technical Commission" and "approved by AO Foundation"
The brands and labels "approved by AO Technical Commission" and "approved by AO Foundation", particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.