Resorbable Tack Fixation System
The system is indicated for fractures of the craniofacial skeleton and reconstructive procedures of the midface and craniofacial skeleton. The resorbable tacks and the existing, approved 1.5mm resorbable screws are very similar in design and size. They are manufactured from the same polymer and in the same processes. Both products are made from 70:30 poly (L/DL)-lactide. The tacks are inserted into predrilled holes in bone to secure resorbable plates and provide rigid fixation. The implants utilize circumferential ribs to provide sufficient retention. Compared to the existing 1.5mm resorbable screws, the tacks provide comparable strength in pull-out and shear over time. The driver grasps the tacks around the head to provide retention during use. During insertion, an intemal shaft automatically disengages the tacks from the instruments and drives the tack head into the countersunk holes of the resorbable plates. Thus, surgical operative time is reduced because the holes are drilled, not tapped, and the tacks are pressed into bone holes rather than threaded in, as the resorbable screws are.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
"approved by AO Technical Commission" and "approved by AO Foundation"
The brands and labels "approved by AO Technical Commission" and "approved by AO Foundation", particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.