Headless Compression Screw (HCS) 3.0 mm
Intraarticular fractures (eg, scaphoid fractures) are best fixed with implants which can be countersunk under the bone surface for protection of the joint surface and adjacent soft tissues. Existing headless screws can be placed below the bone surface but offer only limited control of interfragmentary compression. The 3.0 mm cannulated screw with threaded washer can be used as a lag screw whilst being placed below the bone surface. It provides good interfragmentary compression, but has the disadvantage of a large head diameter which makes it difficult to use in small bones (eg, proximal pole fractures of the scaphoid). The new 3.0 mm Headless Compression Screw has a similar shaft size when compared to 3.0mm cannulated screws, but a smaller head diameter. Unlike other headless screws, the HCS is applied as a lag screw, which allows the surgeon to control closure and the degree of closure of the fracture gap.
The 3.0 mm HCS is indicated for fractures and nonunions, arthrodeses, bunionectomies, and osteotomies of small bones. The use of the Headless Compression Screw in heavily porotic bone is not recommended because of insufficient thread purchase and the likelihood of loss of stable fixation.
The main feature of the HCS system is the control of closure and compression of a fracture gap, which is achieved by means of the compression sleeve. Once the desired compression is achieved, the screw is advanced into the bone using the screwdriver, and the head is sunk.
The leading and trailing threads of the HCS both have the same pitch. This is to ensure that controlled compression can only be applied by means of the compression sleeve. The trailing thread has a double-start to ensure that the compression sleeve attaches to the trailing thread more readily. The cannulation of the screw allows minimally invasive insertion, as well as more accurate placement of the drill bit and, therefore, the screw. The leading thread tip is self-drilling as well as self-cutting. The trailing thread is also self-cutting. The screw is available in stainless steel and titanium. In addition, there are short and long thread screws to provide extra stability in larger bone fragments. The short threaded screws come in lengths from 1030 mm in 1 mm increments and from 3040 mm in 2 mm increments. The long thread screws come in lengths from 1630 mm in 1mm increments and from 3040 mm in 2 mm increments.
Acute scaphoid fracture treated percutaneously with the 3.0 mm HCS
a preop AP
b intraop AP
c 2 wks obl
Case provided by D Campbell, Leeds
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
"approved by AO Technical Commission" and "approved by AO Foundation"
The brands and labels "approved by AO Technical Commission" and "approved by AO Foundation", particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.