LCP Clavicle Hook Plate
The LCP Clavicle Hook Plate is a fixation system with angular stability for both lateral clavicle fractures and acromioclavicular joint injuries. It is indicated for luxation of the AC-joint (Type III, classified by Tossy, or Type III, IV, and V, classified by Rockwood) and lateral shaft fracture of the clavicle (Type II, classified by Neer, or Type II, classified by Jger and Breitner).
The design of the LCP Clavicle Hook Plate includes several new features compared to the existing Clavicle Hook Plate. It implements the LCP concept with elongated combination holes (threaded hole for locking head screws (LHS) and dynamic compression unit (DCU) for cortex screws). Further improvements are a precontoured plate shaft (12 angulation) for better anatomic fit, undercuts, tapered plate end and rounded edges, a hook offset of 5 mm, and tapered hook geometry.
The LCP Clavicle Hook Plate will be provided in four different lengths, with 14 elongated LCP combination holes in the shaft portion. These allow the application of 3.5 mm locking head screws as well as 3.5 mm cortex or 4.0 mm cancellous bone screws. The hook will be offered in three different sizes; 12, 15, and 18 mm.
The new trial implants for intraoperative implant size determination are made out of stainless steel (to be visible under image intensification) and will be offered in a right and left version with three different hook sizes (12, 15, and 18 mm). The hook has the same dimensions as the LCP Clavicle Hook Plate implant. The trial implants are particularly important for use in conjunction with sterile packaged implants.
Fadel GE, Scullion JE (2002). Hook plate fixation for lateral clavicular malunion. AO Dialogue; Volume 15, Issue 1: 14
Concha JM, Illera J, Daza C, Quintero J, Cruz C (2005). Stabilization of Acute Type III AC Joint Dislocations with a Hook Implant. AO Dialogue; Volume 18, Issue 3: 22-27
Case provided by Norbert Sdkamp, Freiburg im Br.
Follow-up x-ray after 10 weeks. Begining ossification in coracoclavicular ligament.
Follow-up x-ray after 5 month. Stable AC joint, full function. Persisting ossification in coracoclavicular ligament without clinical relevance.Follow-up x-ray after 10 weeks. Begining ossification in coracoclavicular ligament.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
"approved by AO Technical Commission" and "approved by AO Foundation"
The brands and labels "approved by AO Technical Commission" and "approved by AO Foundation", particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.