To meet the growing demand for flexible yet strong implants for use craniofacial reconstruction, cosmetic surgery, and repair of craniofacial trauma the new porous polyethylene implants (SynPOR) are well-suited.
They are manufactured from an inert, non-absorbable polymer formulated to contain a network of open and interconnecting pores approximately 100 m 250 m in size. These interconnected pores allow fibrovascular tissue ingrowth and relative host incorporation. The result is fast integration and stabilization, which can be advantageous over the host encapsulation of implants as it can be seen when using implants with a smooth surface.
The material used for these implants is an ultra-high molecular weight polyethylene (UHMWPE), which meets the requirements of ASTM standards and has passed ISO standard tests for biocompatibility. It provides strength and stability where needed but remains flexible. The implants can easily be trimmed and contoured and can be fixated with screws tacks, wire or suture depending on the surgeons method of choice. Final shape modifications can be made in situ.
Among the possible clinical applications are orbital, cranial or facial augmentation and reconstruction cases such as fractures of the orbital floor and wall, cranioplasty, or genioplasty.
In contradiction porous polyethylene implants are not intended for use in the presence of:
- active or latent infection
- inadequate coverage of healthy, vascularized tissue
- full load-bearing applications
- systemic disorders that cause poor wound healing or may lead to soft tissue deterioration over the implant
SynPOR implants are available in 50 x 50 mm sized square shapes in four different thicknesses: 0.45 mm, 0.8 mm, 1.5 mm, and 3.0 mm.
They are provided sterile and pyrogen-free for single patient use. Common cutting and trimming instruments such as scissors, mesh cutters, or scalpels (no electrosurgical devices) can be used for shaping. Loose particles can be removed by rinsing with sterile saline solution. In case a contouring is desired, the sheets can be immersed in heated sterile saline (70 C / 160 F or higher). After a short while the implant can be removed and contoured until the desired form is achieved. The implant will return to the original stiffness when but maintain its new contour after cooling.
Anatomical shapes and incorporated titanium
For the treatment of orbital floor fractures SynPOR implants are also available in anatomical shapes: The guitar pick shaped version comes in 24 x 24mm, 30 x 30 mm, and 35 x 35 mm sizes, each one either 0.8 mm or 1.5 mm in thickness. The fan plates measure 35 mm in radius and are also either 0.8 mm or 1.5 mm thick.
To combine the strength of titanium mesh and the material advantages of porous polyethylene the new SynPOR titanium reinforced sheets consist of commercially pure titanium mesh encapsulated in two layers of UHMWPE. Thus the concern of soft tissue entrapment when placing a titanium mesh implant that has been trimmed to a smaller size, may be reduced. Unlike the other original SynPOR implants, the implants reinforced with titanium allow the surgeon to use an implant that is much thinner to span a large orbital floor defect. In addition, incorporation of the titanium allows for intraoperative and postoperative visualization of implant placement, using standard radiographic imaging.
Hazards and labeling
Due to varying countries’ legal and regulatory approval requirements, consult the appropriate local product labeling for approved intended use of the products described on this website. All devices on this website are approved by the AO Technical Commission. For logistical reasons, these devices may not be available in all countries worldwide at the date of publication.
This work was produced by AO Foundation, Switzerland. All rights reserved by AO Foundation. This publication, including all parts thereof, is legally protected by copyright.
Any use, exploitation or commercialization outside the narrow limits set forth by copyright legislation and the restrictions on use laid out below, without the publisher‘s consent, is illegal and liable to prosecution. This applies in particular to photostat reproduction, copying, scanning or duplication of any kind, translation, preparation of microfilms, electronic data processing, and storage such as making this publication available on Intranet or Internet.
Some of the products, names, instruments, treatments, logos, designs, etc referred to in this publication are also protected by patents, trademarks or by other intellectual property protection laws (eg, “AO” and the AO logo are subject to trademark applications/registrations) even though specific reference to this fact is not always made in the text. Therefore, the appearance of a name, instrument, etc without designation as proprietary is not to be construed as a representation by the publisher that it is in the public domain.
Restrictions on use: The rightful owner of an authorized copy of this work may use it for educational and research purposes only. Single images or illustrations may be copied for research or educational purposes only. The images or illustrations may not be altered in any way and need to carry the following statement of origin “Copyright by AO Foundation, Switzerland”.
Check www.aofoundation.org/disclaimer for more information.
If you have any comments or questions on the articles or the new devices, please do not hesitate to contact us.
"approved by AO Technical Commission" and "approved by AO Foundation"
The brands and labels "approved by AO Technical Commission" and "approved by AO Foundation", particularly "AO" and the AO logo, are AO Foundation's intellectual property and subject to trademark applications and registrations, respectively. The use of these brands and labels is regulated by licensing agreements between AO Foundation and the producers of innovation products obliged to use such labels to declare the products as AO Technical Commission or AO Foundation approved solutions. Any unauthorized or inadequate use of these trademarks may be subject to legal action.